ii Heracell Service Manual – Free download as PDF File .pdf), Text File . txt) or read online for free. incubator heracell SM. HERAcell® CO2 Incubator Operating Instructions. Version: / – 3 -. Contents. 1. Information regarding Safe Operation. Manuals and User Guides for Thermo Scientific Heracell i. We have 1 Thermo Scientific Heracell i manual available for free PDF download: Operating.
||26 December 2005
|PDF File Size:
|ePub File Size:
||Free* [*Free Regsitration Required]
The receptacle for the two device fuses is integral to the socket. All supply connections are installed in the supply interface control box  in the rear of the device. To activate the new settings: Page 70 Note Make sure that power sockets remain freely accessible at all times!
Chapter 1 General Notes Latch, glass door No tools are required to replace the seal. Page If the STOP key is pressed, the routine is interrupted and the contra-con Stop dialog box is opened as a security request.
Page 36 3 additional devices independent of device type The contra-con decontamination routine must then be stopped by pressing the appropriate key. Icon description overview Figure. Actual value CO 8. Table Of Contents The boxes 1 to 4 are delivered with the device. Replacing the gas inlet filter The gas inlet filter CO supply has plastic threads and is screwed by hfracell into the threaded hole at the control box.
Thermo Scientific Heracell 240i Operating Instructions Manual
Mnual in Community Feed. Replacing the door seal The door seal magnetic seal  of the outer door is located in the retaining slot.
To ensure the accessibility for care and maintenance works, keep larger side and rear distances. The temperature displays shows the actual value currently measured in the work space. The support rails  of the shelf system are perforated every 42 mm so that the support hooks  can be inserted variably for any culture container size required.
Reinstall 420i shelf system and the removed components. Press the power switch; the switch illumination goes off.
For devices equipped with the optional gas-tight screen, the measurement opening is located: Wipe out the rests of the water at the floor plate of the water reservoir. Rotate the adjustable device stands using the supplied 24 mm wrench until the shelf is positioned horizontally in all directions.
THERMO SCIENTIFIC HERACELL I OPERATING INSTRUCTIONS MANUAL Pdf Download.
To indicate the start of this routine, the icon overtemperature is displayed. Page 2 If failure is detectec, a special sample protection controlling will be enabled.
CO -incubator in the stack. Navigation through the submenus for device configuration. Let the water run into the bucket  through gravity. Structure Of The Operating Levels The hdracell returns to the Configuration menu.
To avoid overloading the incubator or its shelving be sure to observe the sample weight limits specified in chapter Notice The contra-con decontamination routine manjal be performed every 3 months.
Key for starting the auto-start routine 9. Installing devices without the optional gas monitoring system Gas connection without gas monitoring system Figure.
Page 25 The height of the floor stand should not fall below mm. Mark of prescription devices!
Thermo HERAcell-CO2-Incubator 150i 240i manual en
For any communication with the Technical Support of 240ii Fisher Scientific, please have the device data ready. Chapter 4 Description Of The Device As the data mnaual can contain up to 10, data sets, the data transmission to a PC may take some time.
To set the CO set value: Table Of Contents Gas connection Forums Documents Parts Videos News. Remove all samples an brought in tools from the device, then clean and disinfect them.