GE Healthcare c Anesthesia Machine Technical Reference Manual Software Revision 1.x Approved −12−9 EET Junmin Ji (Rick) M c Product Data Sheet – GE Healthcare c Familiar, reliable, GE – with over years experience in anesthesia and quality leadership? 53 cm/21 in Depth: 40 cm/16 in Side of machine: cm/45 in Height: The c is the ideal solution for customers seeking an affordable, reliable and easy-to-use anaesthesia system. The c is designed and manufactured in.
||16 May 2013
|PDF File Size:
|ePub File Size:
||Free* [*Free Regsitration Required]
Close the cylinder valve and allow the pressure to decay to kPa psi as indicated on the cylinder gauge upper limit of the red band.
GE Healthcare Anesthesia Machine- 9100C, ICU Use, Medical Use
The serial number can be in one of two formats. If ventilator operates from backup battery, check 2 and 3.
Paw Gain Section 4. Perform the checkout procedure Section 3. As pressure decreases, alarms for low O2 supply pressure should occur.
Hold the selector knob with the indicator mark facing down. From the Service Mode, perform the following: PEEP control valve Under high flow conditions, the output pressure should not drop below the specifications in step 6. Close the cylinder valve. Tactile knob indication at: Set the system switch to On and enter the Service Mode. Turn all flow controls fully clockwise zero flow.
Exhalation valve reilef valve test Section 6. The problem is most likely the speaker or control 3. All connections have indexed fittings, filters, and check valves one- way valves.
Pneumatic Vent Engine 1. O2 flush button Zero the pressure gauge Section 5.
The Air and O2 pipeline manifolds include a drive gas connection at the back of the manifold. Robust and reliable, the c is compact and lightweight, easy to manoeuver in any environment. Complete the System Checkout by performing the following steps: Drive gas pushed 900c on the bellows. Inspiratory flow valve GE Clinical Information Systems. Bains Circuit in Chennai.
Tighten the mounting screws evenly to secure the aneshtesia assembly to the pan. Registered under Indian Companies Act The interlock system prevents purging more than one vaporizer at a time.
GE Sustainable Healthcare Solutions
Perform the following calibrations: Remove the four mounting screws from the manifold. Inspect the bellows for damage; replace. Did you just push Flush? If pipeline O2 is not available, open the O2 cylinder valve.
Multipurpose Anaesthesia Machine- 9100C
The way to enter the service mode is: Install the replacement switch assembly: Test the replacement switch assembly: This manual is subject to periodic review, update and revision. Clip the tie wraps 1 from the outlet barb fittings at the side of the switch. Pull out the bag port from the manifold. Verify that T-handles are tight.
Ultrasound designed with you and your patients in mind. Bellows mounting rim loose Remove rim and pop-off valve diaphragm; reseat diaphragm; snap rim into place. Gas supply Pipeline input range: Exit the Service Mode and go to Normal Mode. If test fails, tighten, repair, or replace as needed.