Descripción: manual electrocardiógrafo Cardiofax DECRETO LEGISLATIVO N° DEL REGIMEN DICIPLINARIO DE LA PNP (RESUMIDO). docx. D.L. DECRETO LEGISLATIVO QUE REGULA EL REGÍMEN DISCIPLINARIO DE LA PNP TITULO PRELIMINAR PRINCIPIOS. Grupo Ascenso PNP. LEY DEL RÉGIMEN DISCIPLINARIO DE LA PNP – DECRETO LEGISLATIVO Nº DEL 11 DE DICIEMBRE DE
||Health and Food
||19 February 2009
|PDF File Size:
|ePub File Size:
||Free* [*Free Regsitration Required]
Notification sound volume Select the notification sound volume. Use the number keys on the operation panel to select the number of the desired item. The Setup screen for that item is displayed. Built-in battery One year When the battery deteriorates, the battery operation time is less than 30 minutes after 10 hours charging. The Data management screen appears. Disinfectant Chlorhexidine gluconate solution Benzethonium chloride solution Glutaraldehyde solution Benzalkonium chloride Alkyldiaminoethylglycine hydrochloride 9.
Number Keys On some screens, you can press the panel key to enter a number 0 – 9. Electrostatic energy may generate spike noise which is superimposed on the ECG waveforms. Examples of color-developing materials containing organic solvents: Enter the current date and time with the keyboard on the operation panel.
This may damage the battery. Wash with clean water and see a physician immediately. When a battery is inserted, the electrocardiograph automatically switches to battery power if the power cord is disconnected or decreyo is a sudden power failure. When the provided power cord cannot be used or when equipotential grounding is doubtful such as in a facility with poor groundingoperate the electrocardiograph on battery power.
When it is off, the high cut filter is applied. Format the SD memory card. NOTE Do not lift the electrocardiograph while holding the patient cable and do not bend or pull the patient cable ddcreto. When one minute of ECG waveforms are acquired, recording starts.
Failure to follow this warning may cause electrical shock to the patient and operator. CAL mark The height corresponds to 1 mV waveform amplitude. Potential difference between instruments may cause current to flow to the patient connected to the instruments, resulting in electrical shock micro shock.
Select the age group by pressing the AGE key. If abnormal performance is observed, additional measures may be necessary, such as re-orienting or relocating the ECG Before or after automatic or manual ECG recording, you can record one minute of the rhythm lead.
decreto legislativo 1150 pdf
This warranty does not apply to products that have been modified, disassembled, reinstalled or repaired without Nihon Kohden approval or which have been subjected to neglect or accident, damage due to accident, fire, lightning, vandalism, water or other casualty, improper installation or application, or on which the original identification marks have been removed.
To cancel saving, press the Cancel function key. The lamp lights when copy is available and during printing. Select the default age group when the power is turned 3 – dexreto on. Paper gets jammed in pno recorder. Check if the specified recording paper is used. Action Enter correct values.
EQWEQ by Siete Tacsi Avila on Prezi
After recording, the periodic recording automatically resumes. In automatic recording mode, ECG analysis is automatically performed. Large Polarization Voltage If the leegislativo is still displayed after reattaching the electrode, it is probably due to a large polarization voltage. The battery charge time is at least 10 hours. Examination name Electrode status display area Displays the error message for electrode detachment and noise. Invalid setting file The setting file could not load correctly.
Press any key to turn the screen on again. Otherwise sudden power off occurs when an electrode is detached during recording.
Manual Cardiofax 1150
Display the Data Management – Delete screen. Clean the skin with a ,egislativo moistened with alcohol to remove oil. Select the default sex when the power is turned on. Connect the patient cable to the patient cable connector on the right side panel of the electrocardiograph.
Attach the six electrodes to the cleaned skin site. Change function key increments the value by 1 step. Recharge the battery immediately or operate on AC power. If this happens, turn the power back on to continue using the electrocardiograph. Noise Not enough free memory Overheating.
After the current ECG data file is deleted, the deleting of the remaining files is canceled. The date and time is not correct.
When the patient Legislaativo, gender and age are entered, they are displayed here. If different types are used together, it may cause a large polarization voltage and prevent displaying and recording of the ECG waveforms. Connect the battery cable to the battery pack connector.
The characters are toggled. Select whether to display the age on the Patient Information screen.