Hp B Pdf User Manuals. View online or download Hp B Technical Data Manual. Items are supplied with manuals, accessories and typically a full no-quibble 2 year warranty. Our staff have extensive backgrounds in T&M, totalling over OPERATING AND SERVICE MANUAL PART NUMBER Instruments Covered by this Manual. HP Model A Signal Generator with .
||4 January 2014
|PDF File Size:
|ePub File Size:
||Free* [*Free Regsitration Required]
Press the Done key to exit this menu, and then press manuxl Done key again to return to the Main Software level. Four simultaneous modulation signals, two AM and two FM, may be selected. All Signal Generator settings remain unchanged when the transition from local-to-remote operation occurs.
Set the hgital multimeter as follows: The basic unit for amplitude in the Signal Generator is dBm.
8656B Synthesized Signal Generator, 0.1 to 990 MHz
All fractional digits are truncated. Description SWR is verified by comparing the reflected power frequencies 0. Examples are internal instructions to more than one hardware location, may be serial, parallel, or both and differing amounts of control data and calculations. Changes in these adjustments can mmanual other adjustments, especially level and modulafion accuracies.
Connect the digital multimeter to A13J2 pin 5. Dimensions; H x W x D mm 5.
Directions for Amplitude Offset: Set the digital multimeter as follows; Function Vdc Range 20 V 2. Comments You can save up to complete front-panel set-ups exclusive of increment settings. Frequency Amplitude Modulation 2.
Such maintenance should be performed only by service-trained personnel who are aware of the hazards involved for example, fire and electrical shock.
The implementation of this calibration is different for each mode of Manjal Modulation.
Hewlett-Packard B Signal Generator
A summary of program codes is provided in Tables and This feature is disabled and the original carrier frequency increment 8656g is enabled by pressing the INCR SET key with the frequency function selected. For warranty service or repair, this product must be returned to a service facility designated by HP.
If the amplitude variation is mmanual than 1 dBrepeat steps 4,5 and 6 to compensate the flatness so that the amplitude remains within 0. Read the maximum and minimum power levels from the spectrum analyzer and substitute their value in the following equation and solve for the SWR of the Signal Generator. Refer to Table for the part numbers of tiie power cables and Mains plugs available.
HP 8656B Manuals
All front-panel annunciators and LED segments will momentarily light up for a visual inspection. Press either the [am or FM key.
Hardware execution time will also vary considerably. Amplitude Modulation See Table. Remote Operation Remote Capability.
FM Distortion at Hz and 1 kHz rates: Ideally, the frequency change should be made from the same preset frequency. Alternate models may be suggested for additional features that would make them a better choice in some applications. The Mains plug shall only be inserted in a socket outlet provided with a protective earth contact.
Full text of “HP: B B final”
Areas II through IV make up the hardware execution time. About this Manual We’ve added this manual to the Agilent website in an effort to heip you support your product. Verily that all harmonics are within the specifications shown in the following table.